Facebook Official Page
Share EAGLE library
Have you created EAGLE library?
Did you find an interesting library?
Why not share with the community?
DIY Modules Eagle Library version 1.8.0 has been released.
DIY Modules Eagle Library version 1.7.0 has been released.
DIY Modules Eagle Library version 1.6.0 has been released.
Search for EAGLE Library
ESP8266 LM2596 7805 7 SEGMENT POTENTIOMETER ARDUINO SWITCH RELAY 1N4007 LCD ATMEGA 433 MHZ OLED TO252 BLUETOOTH MODULE ACS712 ARDUINO NANO MOSFET JST POLOLU TRANSFORMER MURATA 315 MHZ PCB ANTENNA BATTERY TL431
Ohter interesting searches:
10K431 BLUETOOTH BF INDUCTOR 400V CAPACITOR RN4020 ARDUINO MEGA RN4020 ACDC 11F 420 433 MHZ PCB ANTENNA INDUCTOR 1N TEXAS NRF24L01 MICROCHIP TDA2005 DIODE CC2500 CURRENT TRANSFORMER 10K431 FINDER MCP TDA RASPBERRY
EAGLE Library Object
|Uploaded by:||Jan Kandziora|
|Download options:||Download original file|
( 44.5 KiB )Download file converted
to the latest EAGLE format
( 101.0 KiB )
|Description:||Temperature Datalogger with 8KB Datalog Memory
This chip combines the core functions of a fully featured datalogger in a single chip. It includes a temperature sensor, realtime clock (RTC), memory, Onewire interface, and SPI interface for connection of an analog-to-digital converter (ADC), as well as control circuitry for a charge pump.The ADC and the charge pump are peripherals that can be added to build application-specific dataloggers. A total of 8192 byte or 4096 word precision readings taken at equidistant intervals ranging from 1s to 273 hrs can be stored automatically.Notes: The datalogger has both a free-run and an operate-until-full mode, which can be selected by software. A 2.8..5V (battery powered) Vdd has to be applied, as all the circuit but the Onewire part of the chip is operated from that supply. This chip is designed to operate together with an MAX1086..1089 A/D converter and a MAX619 chargepump regulator. See the datasheet for details. The TEST_SPLY and TEST_RX pin must be connected to GND, the TEST_CG pin must be left unconnected for normal operation. Author: Orginal device and description from Giovanni Varasano <firstname.lastname@example.org>
Improved by Jan Kandziora <email@example.com>